Glove Box Every Day Routine Maintenance Manual
1) Fill in the use file in time following operating the glove box, which includes "user, drinking water oxygen price, what to do"
two) When changing fuel cylinders in the glove box, the stress is set to the default benefit ± 3mbar
3) The day by day working tension is established according to the running ease and comfort. When not in use for a extended time, the tension is established to one-6mbar
four) Routinely test the wear of the gloves, verify the oil degree of the vacuum pump (between the upstream and downstream markings), and include the pump oil correctly
5) Cleanse the drinking water probe the moment each and every two to three months to make sure precise detection. (Also need to refer to the genuine scenario of the surroundings in diverse user bins
6) If you generally use natural and organic solvents in the glove box, it is recommended that you often substitute the natural solvent adsorbent material, not later than one year at the hottest
seven) Switch the filter component routinely, no later than two yrs
8) In accordance to the real use of your glove box, come to a decision to change the vacuum pump oil cycle, these kinds of as black and innovative technology glovebox yellow pump oil, drinking water or additional natural solvents in the pump oil. The oil replacement cycle should not exceed 2 several years at the latest
9) If a significant total of natural solvent is inevitably uncovered in the course of the experiment, the cycle desires to be stopped for the experiment. Soon after the experiment, to start with thoroughly clean and then start the cycle
10) When the atmosphere in the box is poisonous or damaging gasoline, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and the exhaust port of the cleaning valve need to be fatigued to the outdoor with an exhaust pipe (when exhaust is not allowed to be outdoor, the exhaust gasoline therapy ought to be completed first).